|Year : 2022 | Volume
| Issue : 1 | Page : 44-46
Abdominal twitches due to ICD lead dislodgement; A case report
Yusuf Cemil Oktay1, Muhammed Rıdvan Ersoysal2, Burcin Akgun1, Gorkem Kus2, Engin Deniz Arslan1
1 Department of Emergency Medicine, Health Sciences University Antalya Training and Research Hospital, Antalya, Turkey
2 Department of Cardiology, Health Sciences University Antalya Training and Research Hospital, Antalya, Turkey
|Date of Submission||02-Apr-2021|
|Date of Decision||31-May-2021|
|Date of Acceptance||04-Jun-2021|
|Date of Web Publication||20-Jan-2022|
Dr. Engin Deniz Arslan
Department of Emergency Medicine, Health Sciences University Antalya Training and Research Hospital, Antalya
Source of Support: None, Conflict of Interest: None
Malfunctions of implantable cardiac devices have the potential to be catastrophic. A 63-year-old patient was admitted to our emergency department with new-onset abdominal pain and sensation of twitching on his abdominal wall. One month ago, an implantable cardioverter-defibrillator (ICD) was implanted in him to prevent malignant ventricular arrhythmias. An electrocardiogram was obtained that showed pace spikes unrelated to QRS complexes. The patient was diagnosed with ICD lead dislodgement and the lead was repositioned in another procedure. In conclusion, emergency physicians should be familiar with the problems of implantable cardiac devices.
Keywords: Abdominal twitch, implantable cardioverter-defibrillation, lead dislodgement, reel syndrome
|How to cite this article:|
Oktay YC, Ersoysal MR, Akgun B, Kus G, Arslan ED. Abdominal twitches due to ICD lead dislodgement; A case report. Turk J Emerg Med 2022;22:44-6
| Introduction|| |
Implantable cardiac electronic device lead dislodgement carries significant comorbidities such as malfunctions of implantable cardiac devices and may result in inappropriate therapy such as the paucity of shock delivery in ventricular tachycardia or lack of pace in severe bradycardia. The incidence of implantable cardioverter-defibrillator (ICD) lead dislodgement was found 1.8% in the 1st year of their implantation. In this paper, we present a case with abdominal complaints due to ICD lead dislodgement.
| Case Report|| |
A 63-year-old patient was admitted to our emergency department with new-onset abdominal pain and sensation of twitching on his abdominal wall. At his past medical history, the patient had coronary artery bypass grafting operation, ischemic stroke, and congestive heart failure. The patient's ejection fraction was 30%, and therefore, an ICD was implanted to prevent malignant ventricular arrhythmias 1 month ago. He was on his own medications regularly. He denied any recent trauma to his abdomen or thorax. His vital findings were in normal limits and his physical examination was unremarkable except observable twitching movements on his abdominal wall. An electrocardiogram (ECG) was obtained from the patient that showed sinus rhythm, right bundle branch block (RBBB) with left posterior fascicular block (LPFB), and pace spikes unrelated with QRS complexes [Figure 1]. The patient was monitored and observed that the pace spikes occur simultaneously with abdominal pulsations. The patient was admitted and another operation was performed for the repositioning of the lead. A fluoroscopic imagining was obtained before the procedure that showed the dislodgement of the ICD lead [Figure 2]. During the procedure, coiling of the lead around the generator was observed and the lead of the ICD was uncoiled and repositioned under fluoroscopy. The patient's complaint had resolved after operation, an ECG was obtained that showed sinus rhythm with RBBB and LPFB without any pace spikes and chest X-ray that confirmed normal placement of the lead [Figure 3] and the hospital course was uneventful, and the patient was discharged the day after the procedure.
|Figure 1: Patients' electrocardiogram on admission shows sinus rhythm, right bundle branch block, left posterior fascicular block, and regular pace spikes unrelated with QRS complexes. Note that hidden pace spikes in the third and eleventh QRS complexes|
Click here to view
Written approval was obtained from the patient to publish his case and images.
| Discussion|| |
In this paper, we present a case who had an inserted ICD and abdominal twitches simultaneously with pace activity. As the ICDs have pacemaker (PM) properties and generate pulses below the set limits of heart rate or in case of asystole, in our case, the device did not sense the ventricular systole because of the dislodgement of the lead from the right ventricle and generated paces that innervate nervous phrenicus, the reason of the abdominal twitching.
PMs and ICDs have similar components and they are composed of a power source, electronic circuit, and leads. For both the devices, three types of lead dislodgement were defined in the literature, but naturally, these dislodgement syndromes were reported earlier for PM s then ICDs: Twiddler's, Reel, and Ratchet syndromes.,,,, In Twiddler's syndrome, the first reported lead dislodgement syndrome, the lead of the PM and ICD twists multiple times due to the rotation of PM generator on its long axis.,,,
In 1999, Carnero-Varo et al. reported another form of lead dislodgement and called Reel syndrome in which the PM generator rotates on its transverse axis so the leads coil up around the generator, as in our case., As the components of the PMs and ICDs are similar Reel syndrome related with ICDs might exist but to our knowledge, there were two Reel syndromes reported in the literature in this manner.,
In Ratchet syndrome, the lead of the devices moves toward to generator through the sewing sleeve until the tip of the lead reaches to the pocket.
Age, obesity, proximity to the shoulder joint, hyperabduction of the arm, loose suturing fixation, and pocket mismatch and manipulation of the pulse generator are some important factors that play role in the dislodgement of the leads. In our case, age and hyperabduction of the arm may play some role due to the laxity of tissues for the dislodgement of the lead, although this mechanism is more commonly responsible for the development of Ratchet syndrome. In any case, training the patient for avoiding the manipulation of the device and restriction of excessive movements of the extremity in the early phase of implantation should be prescribed to prevent migration of the leads.
All types of lead dislodgement in all types of implantable cardiac devices may result in extracardiac stimulation, inappropriate shock, undetermined malignant ventricular arrythmias, syncope or asystole which may end in poor outcome. However, our patient had twitching movements due to PM activity of the ICD and therefore presented earlier which probably prevented him from a catastrophic ending.
| Conclusion|| |
Participants who have an implanted cardiac device should be followed up closer, particularly in the first few months, and emergency physicians should be familiar with the clinical scenarios that are caused by the malfunction of these implantable cardiac devices.
Author contributions statement
YCO: Conceptualization, writing original draft; MRE: Writing-review and editing, visualization; BA: Writing original draft, Visualization; GK: Conceptualization, writing review and editing; EDA: Conceptualization, Writing-review and editing, Supervision.
Conflicts of interest
Consent to participate
The authors certify that they have obtained all appropriate patient consent forms. In the form, the patient has given his consent for his images and other clinical information to be reported in the journal. The patient understand that name and initials will not be published and due efforts will be made to conceal identity, but anonymity cannot be guaranteed.
| References|| |
Ghani A, Delnoy PP, Ramdat Misier AR, Smit JJ, Adiyaman A, Ottervanger JP, et al.
Incidence of lead dislodgement, malfunction and perforation during the first year following device implantation. Neth Heart J 2014;22:286-91.
Bayliss CE, Beanlands DS, Baird RJ. The Pacemaker-Twiddler's syndrome: A new complication of implantable transvenous pacemakers. Can Med Assoc J 1968;99:371-3.
Singh G, Khemani H, Khan Z, Bansal NO. Pacemaker Twiddler syndrome – A rare but serious cause of pacemaker malfunction. Int J Clin Cardiol 2018;5:123.
Carnero-Varo A, Pérez-Paredes M, Ruiz-Ros JA, Giménez-Cervantes D, Martínez-Corbalán FR, Cubero-López T, et al. “Reel Syndrome”: A new form of Twiddler's syndrome? Circulation. 1999 Aug 24;100(8):e45-6.
Munawar M, Munawar DL, Basalamah F, Pambudi J. Reel syndrome: A variant form of Twiddler's syndrome. J Arrhythm 2011;4:338-4.
Von Bergen NH, Atkins DL, Gingerich JC, Law IH. “Ratchet” syndrome, another etiology for pacemaker lead dislodgement: A case report. Heart Rhythm 2007;4:788-9.
Asbach S, Brunner M, Zehender M, Bode C, Faber TS. Multiple inappropriate defibrillator discharges due to Twiddler's syndrome. Wien Klin Wochenschr 2005;117:801.
Gungor H, Duygu H, Yildiz BS, Gul I, Zoghi M, Akin M. Twiddler syndrome as a rare cause of implantable cardioverter defibrillator malfunction. J Cardiovasc Med (Hagerstown) 2009;10:352-3.
El-Zein RS, Shah A, Luli J, Stelzer M, Fu E, Amin AK. Lead Macrodislodgement of a Subcutaneous Implantable Cardioverter-Defibrillator Results in a Reel Problem. JACC Case Rep. 2021 Mar 17;3(3):523-527.
Calcaianu M, Bresson D, Lévy J. A potentially lethal dysfunction of a subcutaneous implantable cardiac defibrillator: “The Phantom Menace”? Can J Cardiol 2017;33:1066.e9-11.
[Figure 1], [Figure 2], [Figure 3]